53A-270 ARBITRARY SERIAL PATTERN GENERATOR CARD
53A-270 – Arbitrary Serial Pattern Generator Card
OPT 49 – Differential Output
53A-734 – Analog Cable. or
53A-780 – Hooded Connector
• 10 MHz
• Generates arbitrary serial data patterns at TTL levels with up to 1600 user-defined time durations
• Two independent, programmable output channels
• Time duration values can be updated “on-the-fly”
• Operates in either normal or breakpoint modes
The 53A-270 Arbitrary Serial Pattern Generator (ASPG) provides two independent programmable output channels for generating arbitrary serial data patterns at TTL levels. A memory associated with each channel provides storage for up to 1,600 user-defined time duration values that control the time between high-to-low or low-to-high output waveform transitions. Time duration values can be initially programmed, updated “on-the-fly” while transmitting, or both, including the update of a given time duration while it is occurring. In addition to the time duration values, the user has full program control over the time resolution, initial output state (high or low), and the trigger conditions (software or external trigger control) for each channel.
The basic resolution and accuracy between data transitions is 100 nanoseconds. Four programmable time duration ranges are available. Multiple cards and/or channels can be configured to operate in parallel.
Two modes of operation are provided: normal, and breakpoint. In normal mode, the user specifies the time duration list (1 to 1,600 entries) to be output and the number of times the pattern is to be repeated (1 to 63 or continuous). In breakpoint mode, the user can also specify that a breakpoint be associated with any given transition. Either mode can be operated with internal triggering or external triggering. In normal mode, the output data string repeats, without interruption, the specified number of times. If a subsequent trigger is received while the transmission is occurring, the output data string starts over at the beginning of the time duration list. In breakpoint mode, the Card is retriggered after each specified breakpoint.
The 53A-270 can generate an interrupt to the system controller when a transmission sequence is completed, or at each selected breakpoint in the breakpoint mode.
Built-In-Test Equipment (“BITE”) for the ASPG is provided by a series of fourteen LEDs that provide information on the status, mode, and resolution of each channel.
Number of Channels: Two.
Type of Output: TTL.
Number of Sequential Output Durations: One to 1,600 per channel.
Number of Data Repetitions: One to 63, or Continuous.
Significant Time Duration Digits: Six Decimal Digits (20 bits).
Trigger Modes: Normal Mode, output waveform 1 to 63 times or continuously.
Breakpoint Mode, output memory until the next breakpoint.
Time Duration Between Output Transitions (programmable for each channel):
Internal (software) or external.
External Trigger Input:
Trigger occurs on high-to-low TTL transition.
3,200 24-bit words (1,600 words per channel).
External Clock Input:
Frequency, 1 KHz to 1 0 MHz.
Loading, 1.25 TTL load equivalent.
Serial Pattern, both true and inverted for both channels.
Transmission Active, both true and inverted for both channels.
Internal Clock, buffered, 10 MHz.
OPT 49, differential output.